The 3D Vision system used during the experimentation was the prototype 3D-Optolab, and the CMM was the Zeiss Prismo Vast 7D, equipped with the software Holos, installed at the DIMEG Metrological Laboratory. Both are shown in Fig. 1. The proposed methodology does not foresee the physical integration of the two sensors; instead, their combination at the level of the measurement information is carried out, in a module for the intelligent aggregation of the information coming from the sensors.
Fig. 2 schematically presents the method. The starting point is the acquisition of a number of clouds of points using 3D-Optolab. These are then imported into the CAD environment PRO/ENGINEER. The initial ‘rough’ CAD model of the surface is obtained by using the modules available in the CAD environment PRO/E. This model is used to ‘feed’ the CMM in the contact, accurate digitisation step. The a-prori knowledge of a ‘rough’ description of the surface allows an efficient programming of the scanning and digitising path, and reduces the number of touch points and of the iterations needed to achieve the complete difitisation of the object. The methods was tested on a number of objects. The experimental results are presented and discussed in the References.
This research activity has been further developed in the frame of the project “Development of a novel methodology for the reverse engineering of complex, free-form surfaces, combining three-dimensional vision systems and Coordinate Measuring Machines” funded by the Italian Ministry of Research, in 2000. Two further Laboratories participate to this project. These are the DIMEG Metrological Laboratory, University of Padova, and the 3D Vision Group located at the Dipartimento di Elettronica e Informatica, located at the Politecnic of Milan. The objectives of this work are well described by the scheme in Fig.3.
(i) development of a reliable, easy to use, optical digitiser, able to generate 3-D point clouds that describe various parts of the object, each from a specific viewpoint; the measurement system, should be easily movable in space, in order to be able to “observe” the target object from different perspectives, and to create a set of point clouds that completely describe the object itself;
(ii) development of procedures for the registration of the point clouds;
(iii) development of the procedures for the creation, starting from the registered views, of 3D models of the shapes;
(iv) metrological validation of the models by means of the CMM.
The second purpose of the project (‘Optical/Contact RE in the Figure) is to optimise the RE process from the viewpoint of the accuracy of the representation of the object, without increasing the process time. The approach is close to that one represented in Fig.2; however, the initial representation of the CAD model has to be obtained starting from the 3D models above mentioned.
The third purpose of the project (‘Metrological validation of point clouds’ in the Figure) is closely related to the activity aimed at metrologically validating, by means of the CMM, the point clouds generated by the optical digitiser.
The final goal of the project (‘Validation for RP’ in the Figure) is the verification of the suitability of the 3D models for the Rapid Prototyping process.
The activity carried out by our Group resulted in two research products. The former is the optical digitiser OPL-3D. The design and the development of the instrument have been completely performed by the Group. The metrological characterization has been performed in collaboration with the Laboratory located in Padova.
The latter is a suite of software tools for the alignement of the point clouds in the multi-view acquisition process. These tools perform, in a semi-automatic way, the estimate of the rototranslation matrix between pairs of point clouds. A further improvement is performed by the research Laboratory located in Milan, basically aimed at achieving a completely automatic process. The Software page of this site reports more details about the procedures.
V. Carbone, M. Carocci, E. Savio, G. Sansoni, L. De Chiffre, “Combination of a vision system and a Coordinate Measuring Machine for the Reverse Engineering of Freeform Surfaces”, Int. J. Adv. Manuf. Tech., No. 17, 263-271, 2001.
G. Sansoni, A. Patrioli. “Combination of optical and mechanical digitizers for use of reverse engineering of CAD models”, Proc. of ODIMAPIII, Optoelectronic Distance Measurements and Applications, 301-306, Pavia, 2001.
G. Sansoni, M. Carocci, “Integration of a 3D vision sensor and a CMM for reverse engineering applications”, Italy-Canada Workshop on 3D Digital Imaging and Modeling Applications of Heritage, Industry, medcine & Land, Padova, Italy, April 2001.
G. Sansoni, S. Carmignato, E. Savio, “Validation of the Measurement Performance of a Three-Dimensional Vision Sensor by Means of a Coordinate Measuring Machine”, Proc. IMTC 2004, Vol 1, pp. 773-778, Como, May 2004.